Service release of GMG ProofControl 184.108.40.206
GMG ProofControl 220.127.116.11 has been released today, Thursday, June 22, 2017.
ProofControl 2.0 - 2.2 ➝ 2.3 New license required, chargeable
ProofControl 2.3 ➝ 18.104.22.168 No new license required
• Windows 10
• Windows 8.1
• Windows 7
• Windows Server 2012 R2
• Windows Server 2008 R2
GMG ProofControl 2.3
1. Close all GMG applications.
2. Extract the file GMG_ProofControl 2.3.zip.
3. Double click the setup file Setup.exe to run the installation.
The following technical issues have been resolved in ProofControl 22.214.171.124. Please note that the list describes the problem, not the solution.
Language reverts to English on change of Measurement mode
The language of ProofControl changed from German to English after changing the measurement mode, e.g. from M1 to M0. [70806, 70853, 71891, 71877, 72045, 73236]
Measured values not saved
In a few cases the measured values were not saved in ProofControl. [72766, 72849]
Connection to measurement device lost
After switching the measurement condition ProofControl sometimes remained in the status "Measuring device is initialized...", whereas there was a pop-up window saying: "Measuring device is not connected". In some cases, the options greyed out and the software did not react anymore after changing the measurement condition. [72766, 72849]
IDEAlliance2013 – Custom ProofStandard
Jobs with custom ProofStandards and IDEAlliance2013 control strip did not appear in the ProofControl job list. 
Verification failed for Targets containing 0.00 values
The Delta-CMC could not be calculated when the target LAB contained a 0.00 value (e.g. target for A23 = LAB 16/0/0). ProofControl showed "NaN" instead of verification results. 
Orientation of control strip on the i1iO table
The X-Rite i1i0 could not read media wedges, which were rotated 180°C (C24 Patch in the upper left corner and A1 in the right corner). 
PrintStandard list not sortable when language is not English
The list of PrintStandards were only sortable by clicking on the column header when the language was English. When the language was other than English, sorting did not work. 
ProofStandard not consistent with PrintStandard list
The ProofStandard in the job list was not identical to the PrintStandard in the drop-down list of ProofControl. 
ProofControl Inline labels printed on Zebra printer
This happened in mixed environments, in which some jobs are created to be measured with EyeOne in ProofControl, other jobs are measured and verified with the spectroproofer (PCI). Instead of only adding the ProofControl Inline jobs in the ProofControl history, the results were printed on the Zebra printer. 
Column width change had impact on PrintStandard order
When the column width of the PrintStandard list was adjusted, the PrintStandard order changed. 
New ISO 12647-7:2016
This version contains several improvements related to the new ISO 12647-7:2016 standard.
End of life
Please note, that we will no longer provide bug fixes, maintenance releases, workarounds, or patches for bugs found in GMG ProofControl 126.96.36.199 or older.
GMG ProofControl 188.8.131.52